A major advantage of SPR technology is that the detection does not require the analyte to be labeled, such as with a fluorescent molecule or a redox-active tag. This is because SPR directly detects changes in refractive index resulting from changes in mass at the sensor chip surface. For researchers interested in pharmacology and pharmacokinetics or in general pharmaceutical research or drug discovery, this capability of label-free detection is particularly attractive.
In this application note, the first example (Figure 1) shows binding of acetazolamide to carbonic anhydrase II (CAII), which is an enzyme that rapidly converts carbon dioxide and water to bicarbonate (with proton as a co-product). Carbonic anhydrase inhibitors are a class of pharmaceuticals that inhibit the activity of carbonic anhydrases. Clinically, these inhibitors have been used as antiglaucoma agents to alleviate mountain sickness and to manage neurological disorders. Acetazolamide (molecular weight = 222.22 Da) is one of the commonly used carbonic anhydrase inhibitors. The binding rate constants and affinity value obtained from the kinetics analysis (orange lines) of the SPR binding curves (black lines) are shown in the inset of Figure 1.
The second application (Figure 2) is concerned with the detection of triclosan binding to protein InhA, the enoyl-acyl carrier protein reductase that is essential to the bacterial Mycobacterium tuberculosis, which is a causative agent for tuberculosis. InhA is involved in the biosynthesis of type II fatty acid. Triclosan (molecular weight = 289.54 Da) is a small molecule that has antifungal and anitbacterial effects. As can be seen, the attachment of triclosan to the InhA protein preimmobilized onto a CM dextran chip can be readily detected by SPR. Compared to the data in Figure 1, the binding affinity, KD, is more than an order of magnitude weaker, suggesting that acetazolamide binding to CAII is stronger than triclosan to InhA. As can be seen from both figures, excellent fits were achieved, and the kinetic and binding parameters in the table in good agreement with published results. The feasibility for small molecule detection can be attributed to the high sensitivity and low noise of BI SPR detection technology, the fine precision and stability of its BI-DirectFlow™ sampler delivery system, and the excellent performance of BI sensor chips having high loading capacity with low non-specific adsorption.
The two examples described in this note demonstrate that SPR can be used to study the binding interactions of small molecules to proteins. SPR for small molecule detection is a powerful technique that can be used in applications such as screening drug candidates to combat cancers, alleviate the symptom of neurological disorders, and discover antifungul and antibacterial agents.
Download a PDF of Application Note 119: Small Molecule Detection by Surface Plasmon Resonance (SPR)
- Supuran CT, Scozzafava A, Casini A (March 2003). "Carbonic anhydrase inhibitors". Med Res Rev 23 (2): 146–89